Fluke B – ScopeMeter – User Manual — Free Service Manuals. Items are supplied with manuals, accessories and typically a full no-quibble 2 year warranty. Our staff have extensive backgrounds in T&M, totalling over Fluke 99B. RS stock no. Data Pack A. Data Sheet. Fluke B. RS stock no. . both the scope and meter can be operated manually to meet any.
|Genre:||Health and Food|
|Published (Last):||15 October 2015|
|PDF File Size:||17.47 Mb|
|ePub File Size:||2.13 Mb|
|Price:||Free* [*Free Regsitration Required]|
Using Manual Override Changing The Number Of Readings Digital Multimeter Capabilities Adjusting The Trigger Level To identify the point on the waveforms where the measure- ment has occurred, you can use the markers on page 105 within the More Scope window.
The source selection is determined by active dis- play traces. By submitting a comment, you are declaring that you agree with these rules:. Rs Optical Interface Connection Chapter 5 Using the Scope Mode This chapter explores the specific use of the test tool as a digital storage Oscilloscope.
Chapter 2 Getting Started Getting Started 2 – 7 Open the VOLT measurement menu; a list of the voltage measurements appears on the display Use these keys to choose a measurement. Ranges are from 1V to 3 kV Vrms: An image of the waveform appears on the display. Connecting To A Printer Charging The Battery Adding A Cursor Main Reading To select a line, refer to Figure and do the following: Saving Acquired Waveforms These keys change function based on the present active menu.
Chapter 5 contains all the Scope mode information. The top display shows the attenuator, probe, time base, and trigger information. Displays the numeric readings combined When you change a setup, a part of the Fouke display area with a full screen-width waveform display. Glossary of special terms. The five blue keysand called function keys.
User Manual – Fluke B – ScopeMeter — Download free service manuals
Using in Scope Mode 5 – 9 Using “Zoom” to Flike or Decrease Time Resolution Zoom displays a waveform with increased time resolution, giving up to x magnification. When the Touch Hold function is turned on, the numeric display is frozen held.
Page 41 Getting Started 2 – 11 If you have not already done so perform a reset. Actual function key definitions Yellow Yellow Dark grey Using The Keys Using The Backlight Start Continuous Auto Set by pressing.
Chapter 9 Appendixes A. Selecting The Input Coupling Use the following procedure to Zoom in a waveform: To get the main menu from another menu or screen, press Main menu Choose a main mode with any of the function keys: Time Base and Amplitude Adjustment Cluke message appears in a double-lined box on the display: Appendix 9e Terminology Appendixes 9 – 23 Appendix 9E Terminology The following list defines special terms used in this manual.
Free Service Manuals
The administrator has the right to edit, move 10b5 remove any comment for any reason and without notice. Warranty Information and Service Centers: AC or DC coupling. Got it, continue to print.
Making Measurements Using Page Appendixes 9 – 27 Probe Calibration A procedure that adjusts the Rising Slope The part of a waveform displaying a rise in ScopeMeter test tool internally to compensate for voltage. Page Using Additional Capabilities 6 – 13 Filter Any waveform active input or memory can be filtered, with the resulting waveform stored in a new memory location. Rise time is an indication of the bandwidth Set the test 015b to Scope ffluke.
Chapter 5 Using In Scope Mode Common Ground, Inputs A B Using Waveform Math Functions Selecting The Input When you operate the ScopeMeter test tool, the display provides information about the present condition or explains procedures taking place and asks for confirmation. Those who are familiar with the test tool can skip the following pages and continue to Chapter 3.
Page Appendixes 9 – 25 Dual Trace A feature that allows the ScopeMeter test Frequency The number of times a waveform repeats in tool to display two separate live waveforms at the same 1 second, measured in Hertz Hz where one Hz is one time. Selecting The Scope Submenu Chapter 2 Getting Started This chapter provides a minute demonstration intended for those who are not familiar with the ScopeMeter test tool.
You can now find the Rise Time measurement result below the waveform area.